Download HP A RF IMPEDANCE-ANALYZER OPERATION SM service manual & repair info for electronics experts. The HP A RF Impedance Analyzer measures 14 parameters Parameter measured: IZI, Swept measurement: manual or automatic sweep from start to. Agilent A Test Fixture. Operation and Service Manual. Third Edition. Agilent Part No. February Printed in Japan.
|Published (Last):||13 June 2012|
|PDF File Size:||18.71 Mb|
|ePub File Size:||6.9 Mb|
|Price:||Free* [*Free Regsitration Required]|
Sorry, that last line of code there is supposed to read: I am able to communicate with the device through the GPIB interface, and am also able to retrieve information.
IIRC, the sweep parameters were something like this: I went through the manual before, but when I saw their GPIB section, the strings were so convoluted and so unlike anything I’ve ever seen before that I assumed it was referring to something different.
I think the would probably take the old commands, but ph4191a ALSO has new commands. Thanks for your help guys. Good luck with it Matt – you’re right, the commands for the A are completely different from the – probably because of its age.
If not, what would creating a driver for it entail?
A RF IMPEDANCE ANL [Obsolete] | Keysight (formerly Agilent’s Electronic Measurement)
Message 6 of 6. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. Accepted by topic author Matt C.
manuwl Has anyone done anything on interfacing with the A since Matt C’s last post? When I gave it a try, though, it worked fine. Would you mind posting the code you used on the A?
The forums emoticons took over for a bit there.
4191A RF IMPEDANCE ANL
Message 1 of 6. Most Active Software Boards: Message 3 of 6. In particular, after setting up and initiating the instrument, I am initiating the measurement and reading it as follows:.
In particular, after setting up and initiating the instrument, I am initiating the measurement and reading it as follows: You really need the manual. Mwnual 5 of 6.
Agilent / HP 4191A RF Impedance Analyzer
I have worked with the A and, as I recall, the commands are nothing like any other instrument I have seen. Should I be looking at doing it with LabView or something else?
In the code above, MHz is an arbitrary frequency. I have not worked with that instrument, but the Agilent web site has the operating and service manual here. It’s 84MB, so I hope you have a fast link.
Guess you never know until you try! However, rather than being returned the data at MHz, I am returned the data at whatever frequency the analyzer is currently set at.
HP A for Sale|Impedance Analyzers|Impedance Measurement Products|Test Equipment Center
Message 2 of 6. I suspect that this difference might be causing my problems. Message 4 of 6.